ProductsMeter solution Lancing tools Data management Apps & software regulating Diabetes SUPPORTMeter equipment
The rewildtv.com Performa blood glucose meter makes experimentation your blood street quick and also easy. It’s ready to use ideal out the the box, requiring no set-up. Every meter offers rewildtv.com Performa test strips and an rewildtv.com lancing device. Advantageous alerts repeat you to check before and also after meals, providing you an useful information to assistance therapy decisions for a healthy active life.
You are watching: Accu-chek e-9
After 2 minutes undisturbed with strip inserted, or if it is provided on yet left undisturbed because that 2 minutes
18 month after production date; test strips remain stable approximately expiry day printed on test strip vial, also after opening (test piece container have to be closeup of the door tightly after every test piece is removed)
How to use the rewildtv.com Performa
Get exact results in 3 steps:Insert her test strip into the meter and also it turns on automatically.Apply your blood sample come the easy-to-see yellow window. Only 1 small drop is needed.In about 5 seconds, your test results will appear in the easy-to-read display.
This website includes information on commodities which are targeted come a wide range of audiences and could save on computer product details or information otherwise not easily accessible or precious in your country. You re welcome be mindful that we execute not take any kind of responsibility because that accessing such details which might not comply with any kind of valid legal process, regulation, registration or intake in the nation of her origin.
Battery is dead. Insert brand-new battery. Insert one 3-volt lithium battery (coin cell type CR2032)Display is damaged. Contact (insert affiliate contact information)Meter is defective. Contact (insert affiliate call information)Extreme temperatures. Move the meter come a an ext temperate area.
Your meter steps blood glucose in ~ a defined selection (0.6–33.3 mmol/L).The display may indicate that your blood glucose is above than the measuring range of the device (33.3 mmol/L).Immediately repeat the test. If her test result matches just how you feel, call your healthcare professional immediately.If this does not enhance how friend feel, contact.
Your meter procedures blood glucose in ~ a defined variety (0.6–33.3 mmol/L).The display screen may show that her blood glucose is lower than the measuring range of the mechanism (0.6 mmol/L).Immediately repeat the test. If her test result matches how you feel, call your healthcare expert immediately.If this walk not enhance how you feel, contact.
The check strip may be damaged or not properly inserted. Remove and reinsert the check strip, or change it if damaged.
Your blood glucose might be incredibly high or a meter or a test strip error has actually occurred.If her test result matches how you feel, call your healthcare experienced immediately.If your test an outcome does not match how you feel, repeat the blood glucose test.If the E-3 code still appears for her blood glucose test, your blood glucose result may be very high and over the system’s reading range. Call your healthcare experienced immediately.If the 2nd test an outcome does not match how friend feel, execute a manage test through the control solution and a brand-new test strip.If the control an outcome is in ~ the acceptable range, review the appropriate testing procedure and also repeat the blood glucose test v a brand-new test strip.If the control result is not within the acceptable range, call Roche
Not enough blood or control solution was attracted into the test piece for measure up or was used after the test had started. Discard the test strip and also repeat the blood glucose or manage test.
Blood or regulate solution was used to the test strip prior to the flashing drop symbol appeared on the display. Discard the test strip and repeat the blood glucose or manage test.
An digital error occurred, or in rarely cases, a provided test piece was removed and also reinserted. Turn the meter o. And also on, or take it the battery the end for 20 seconds and reinsert it. Perform a blood glucose or control test.
The temperature is over or listed below the proper variety for the system. Refer to the test piece package insert for mechanism operating conditions. Relocate to one area with the suitable conditions, wait till the meter and test strip reach the operation temperature (8-44 °C), and repeat the blood glucose or control test. Execute not artificially heat or cool the meter.
The battery is practically out of power. Change the battery now. If the blog post reappears ~ the battery has been replaced, remove the battery again, press any type of meter button, then reinsert the battery.
See more: " Picture Puzzle Piece By Shel Silverstein, Picture Puzzle Piece
The time and date settings might be incorrect. Make certain the time and date space correct and also adjust, if necessary.